News about Science Data Product Forum
We would like to draw your attention about the existence of the Science Data Products internet forum which can be found at: http://www.eso.org/sci/data-processing/forum.html
On this page, we list the main changes that affected the instrument and this web site. For news on possible problems see also this QC page.
- 15 November 2014: The FORS2 LADC, which was suffering from a poor coating, was replaced by the one of FORS1 whose coating had been removed. This led to a gain in throughput of 0.1 ( in the blue), 0.07 (in V), 0.05 (in I), and 0.04 (in I) magnitudes, due to a reduction of the scattering.
- 24 August 2014: The FORS keyword HIERARCH ESO INS SLIT WIDTH (slit width in mm) was found wrong in the LSS, HIT and FREE mode, being a factor two too small. This affected all observations done with FORS1 and FORS2 till now. It has been corrected and all files in the archive corrected. The slit width in arcsec is correct.
- 24 June 2014: Since the last CCD exchange on 5 June and 22 June, FORS2 had a rotation misalignment of 0.024 radians. This could affect some runs, so Pis of runs executed during this period should take particular care.
- 3 June 2014: New tests confirm that cross-talk from linear to circular polarisation is present (confirming Bagnulo et al. 2009). Note that it appears only when we want to measure a weak field of circular polarisation in a source that is strongly linearly polarised. It is stronger in the blue (~10%, i.e., a signal of linear polarisation of the order of 10% is seen as a signal of circular polarisation of 1%) than in the red (in the R filter seems about 5%), but is not stable in time. Origin may be the LADC or the SR-collimator. Further tests are ongoing, but users should be cautious in interpreting their FORS2 polarisation data.
- 6 February 2014: The blue calibration lamps have been behaving erratically between 5 and 29 January, leading to underexposed spectroscopic flat fields in the blue (and thereby affecting grisms 300V, 600B and 1200B).
We have retaken when possible new calibrations, so users should check if their data are well calibrated.
- 06 November 2013: The coating of the LADC is severely worn off since 2 years. Programmes that require extremely precise photometry or spectrophotometry over several hours (e.g. transiting exoplanets) should thus not be attempted for now.
- 15 October 2013: Over the last year, we have had several instances where the retarder angle will go to an INDEF position - this will be in the header files (or sometimes nothing is written in the file) - and there is no way to know what was its actual position, although it seems to be mainly a software bug. This affects mostly previous visitor runs, and very seldomly, service mode runs.
- 01 October 2013: calibration OCA rules are now available for historical data, both for FORS1 and FORS2, back to March 2002. For these data, calselector will provide the necessary calibrations.
- 15 February 2013: It was discovered that due to a bug in the associated software, MOS calibrations taken since July 13, 2011 were systematically taken without slit #19. In other words, there has been no FLAT or WAVE taken for this slit since the above mentioned date.
- August 2012: from 10 to 15 of August the instrument was operated with the E2V (blue) mosaic, due to a failure in the control electronics of the MIT (red) mosaic.
- January 2010: the paper by Moehler et al.: Correction of Field Rotator-Induced Flat-Field Systematics has been published.
- April 2009: FORS1 decommissioned
- September 26, 2008: P83 proposers wishing to use grisms not covered bythe Standard Instrument Configuration should be aware that suchrequests will not be considered for Service Mode. For Visitor Modeproposals, users should propose for and base their time request on agrism covered by the Standard Instrument Configuration (found inSection 2.2 of the manual). In addition to this, they should include adescription of the gains that could be made from using a grismpreviously available on FORS (eg. 600V, 1200g). Such requests will beconsidered on a case-by-case basis during the technical evaluation.
- September 1, 2008: There was a error in Table 2.1 ofthe manual showing the "Standard Instrument Configuration". This has now been fixed and the information can also be found from the following link.
- August 29, 2008: Important information from the P83 Call for Proposals - "Firstly, the polarimetric capabilities previously offered on FORS1 will be transferred to FORS2. They will be available on FORS2 for observations during P83. Secondly, the blue-optimized E2V CCD that is currently on FORS1 will be available for Visitor Mode observations with FORS2 during the second half of P83. Finally, the standard instrument configuration for Service Mode observations with FORS2 has changed to incorporate a number of the high-throughput filters and VPH grisms previously used on FORS1". We will update this information as it becomes available during the semester. Users should consult the P83 FORS user manual, available here.
- March 1, 2008: Important extract from the P82 Call for Proposals - "Thepolarimetric capabilities of FORS1 will only be offered during thefirst half of P82. After this time, FORS1 will be offered in itsstandard configuration with the exception of the IPOL and PMOS modes.Due to the arrival of X-Shooter at the Cassegrain focus of UT2, it isexpected that FORS1 operations will end during the second half of P82."
- February 27, 2008: Updated FORS1 standardinstrument configuration - The R_BESS filter is moved to from Wheel 1to Wheel 3 and the I_BESS filter from Wheel 3 to Wheel 1. This meansR_BESS will be available for IPOL observations, but I_BESS will not.
- July 16, 2007: Links to documents relating to the "FORS absolute Photometry Project" added to web page here.
- May 21, 2007: FORS spectroscopic pipeline released. See the DFS pages for more details here.
- April 12, 2007: These webpages are updated to take into account the new blue FORS1 detector.
- April 1 to April 6, 2007: The second commissioning of the new bluesensitive E2V CCD on FORS1 is completed with success. The new cameraand new high throughput broad band filters are then already offered inP79.
- March 1, 2007: The new mosaic CCD will be offeredon FORS1 for the beginning of P80 at the latest, so all P80 proposersshould read the information at Blue upgrade to find details of the enhanced capabilities of the instrument. Inaddition the standard configuration of FORS1 will change from P80 toincorporate the new filters.
- January 30 to February 4, 2007: First commissioning of the new blue sensitive E2V CCD on FORS1
- November 28, 2006 to December 9, 2006: FORS1 annual maintenance (LADC cleaning, new calibration screen).
- November 28, 2006 to December 6, 2006: UT2 M1+M3 mirrors re-coating.
- April 1, 2006 : A new HIT mode, the 'multiple-shift' mode, is available on FORS2 for fast simultaneous spectroscopy of science and reference targets.
- April 1, 2006 : A new high through-put Volume Phased Holographic grating is offered for FORS1. The new 1200B grism covers the range 373-497nm with a dispersion of 0.61Å/pixel and replaces the 1200g grism in the FORS1 standard grism set.
- August 30, 2005 : As of P77, the FORS2 Echelle is decommissioned and replaced by the new and more efficient 1200B grism on FORS1.
- February 25, 2005 : These new FORS webpages are released
- December 13, 2004 : New Rapid Response Mode (RRM) running for both FORS1+2
- June 15, 2004 : New holographic grism GRIS_1200g installed on FORS1
- June 5, 2004 : FORS1 moved to Kueyen (UT2) and FORS2 to Antu (UT1)
- February 27, 2004: Important update on the FORS1 instrumental polarization